The key control parts are
2. is a refrigeration compressor
, 有的用红外线加热 , 或是直接用电阻丝加热 3.Hot fan , some use infrared heating , or directly use resistance wire
The measuring end of the temperature probe extends into the air inside the incubator, and can not be in contact with objects or the cabinet. The temperature in the cabinet is monitored in real time. On the control panel, the constant temperature range of the incubator can be set, that is, the upper limit of the allowed temperature Lower limit. When the probe detects that the temperature is lower than the lower limit, the heating fan is turned on and the temperature starts to rise. When the probe detects that the temperature is higher than the upper limit, the refrigeration compressor is started to cool and the temperature drops.
Some thermostats are more advanced, and you can set the degree of deviation. For example, under normal circumstances, the temperature should start heating when the lower limit is reached, and the heating is later, because the temperature may fall for a period of time after the heating starts, you can set Deviation to make it heat or cool in advance.
t=1.2mm ） A3 钢板数控机床加工成型，外壳表面进行喷塑处理，更显光洁、美观 Outer box material: All are made of high-quality ( t = 1.2mm ) A3 steel plate CNC machine tool. The surface of the outer shell is plastic-sprayed, which is more smooth and beautiful
SUS304 ）板 Inner box material: imported high-grade stainless steel ( SUS304 ) plate
Insulation material: high density glass fiber cotton
The circuit system side adopts door opening to facilitate maintenance and repair
Sealing: Double-layer high-temperature-resistant high-tension sealing strip is used between the door and the cabinet to ensure the tightness of the test area
Door handle: Easy to operate with non-reaction door handle
PU 活动轮 Casters: high quality fixed PU movable wheels at the bottom of the machine