1. Studio size: 400 * 500 * 400mm (depth × width × height);
2. Dimensions: 870 * 1000 * 1700mm (depth × width × height);
3. Temperature range: -20 ° C to + 150 ° C;
4. Temperature fluctuation: ≤ ± 0.5 ℃;
5. Temperature uniformity: ≤1 ℃;
6. Temperature deviation: ≤ ± 2 ℃;
7, heating temperature rate:
From + 20 ～ -70 ℃, about 70min.
From + 20-100 ° C, about 55min.
8. Wind speed: 1.7 to 2.5m / s;
The above indicators are measured at ambient temperature ≤25 ℃, atmospheric pressure, no-load, no-load conditions, 1/6 space from the inner wall of the box;
9. Power: about 3.5Kw;
10. Power supply: 220V ± 10% V; 50Hz;
11. Humidity range: 20% -98% RH (AT + 25 ℃ ～ + 85 ℃).
12. Humidity error: + 2 / -3% RH (above 75% RH), ± 5% RH (below 75% RH).
1. New and perfect arc shape design, high-quality appearance, beautiful appearance, the outer box is treated with high-quality matte line SUS304 # stainless steel plate.
2. The liner is imported (SUS # 304) high-quality mirror stainless steel plate.
3. Insulation material: rigid polyurethane rigid PU foam + high-density glass fiber cotton. The thickness of insulation is 100mm.
4. Wind transport system: using long shaft motor, high and low temperature resistant stainless steel multi-wing impeller, to achieve the intensity of convection vertical diffusion cycle.
5. A double-layer high-temperature-resistant high-tension seal is used between the door and the cabinet to ensure the tightness of the test area.
6, the use of non-reaction door handle, easier operation.
7. The bottom of the machine adopts high-quality fixed PU movable wheels with horizontal feet.
8. The observation window is made of multi-layer hollow tempered glass, and the inside is laminated with a sheet-type conductive film (the test process can be clearly observed).
9, test hole (left side of the machine)-50mm, can be used external test power line or signal line.
10. Comprehensive safety protection device. When any fault occurs, the fault status will be displayed on the screen and the power will be cut off immediately to ensure the safety of the machine itself, the tested product and the user.