Jiaxingcheng thank you for coming
13712718109 吕先生 , 确认相关产品精 量及款式型号等，真诚期待与您合作，谢谢！ Our company is based on the operation mode of factory production + online sales, specializing in the production and design of various electric industrial ovens, high temperature ovens, IR tunnel furnaces, PCV production lines, Uv curing furnaces, fuel injection cabinets, automatic fueling lines, complete sets Painting equipment can be customized according to customer's needs. If you need customization , please contact us 13712718109 Mr. Lu to confirm the precision and style of related products . We sincerely look forward to working with you, thank you!
304#不锈钢、 厚1.5mm； 1.Inner material: 304 # stainless steel, 1.5mm thick;
冷轧钢板，厚1.0mm，粉体烤漆处理； 2. Outer material: cold-rolled steel plate, 1.0mm thick, powder paint treatment;
耐高温石棉绳； 3. Sealing material: high temperature resistant asbestos rope;
PID数显仪表（台湾台松保仪表），单点式控温可自动演算，PV/SV同时显示，按键设定； 4, temperature control: PID digital display instrument (Taiwan Taisongbao instrument), single-point temperature control can be automatically calculated, PV / SV is displayed at the same time, key setting;
固态SSR断电器； 5. Power controller: solid-state SSR circuit breaker;
1秒-9999时（可选择设定小时、分钟、秒）； 6. Timer setting: 1 second-9999 hours (optionally set hours, minutes, seconds);
耐高温不锈钢加热管； 7, heating material: high temperature resistant stainless steel heating tube;
常温+50-500℃ 温度可调，常用450℃以内； 8. Temperature range: normal temperature + 50-500 ℃ temperature adjustable, usually within 450 ℃;
100℃约12分钟左右； 9. Heating rate: about 12 minutes at 100 ℃;
耐高温硅酸铝棉，保温效果好； 10. Thermal insulation material: high temperature resistant aluminum silicate cotton, good thermal insulation effect;
3-32V； 11. The output is: 3-32V;
1℃； 12. Display analysis: 1 ℃;
100℃±2℃，200℃±3℃，300℃±4℃，400℃±5℃，500℃±6℃空臬测试； 13. Uniformity of temperature distribution: 100 ℃ ± 2 ℃, 200 ℃ ± 3 ℃, 300 ℃ ± 4 ℃, 400 ℃ ± 5 ℃, 500 ℃ ± 6 ℃ empty test;
当温度失控时，超过超温保护设定温度时，自动停止加热供电，从而保护产品及机器的安全； 14. Over-temperature protection system: When the temperature is out of control, when the over-temperature protection set temperature is exceeded, the heating and power supply will be automatically stopped to protect the safety of products and machines;
接地保护、快速保险； 15. Circuit protection system: ground protection, fast insurance;
强制送风循环，内箱左面出风，右面吸风；马达为耐温长轴耐温专用型，马达功率37CW/380V；风轮为多翼式涡轮型几轮7.5寸；进风口于机台右侧、排风口于左侧。 16. Air supply system: forced air circulation, the left side of the inner box draws air, and the right side draws air; the motor is a temperature-resistant long-axis temperature-resistant type, the motor power is 37CW / 380V; ; The air inlet is on the right side of the machine and the air outlet is on the left.
Notice to customers
Product pictures, prices, and product attributes are for reference only; specific product requirements are subject to customer requirements, and prices are subject to offline quotation plans. All product prices in this shop are not transaction prices and are for reference only! Welcome to inquire or negotiate in the factory!
所有商品我们在发货前有专门质检部门进行严格检查，确保无任何质量问题后，方可发货 Product problems: We have a special quality inspection department to strictly check all goods before delivery, and ensure that there are no quality problems before we can ship
Product pictures: The pictures are taken 100% in kind, but some products may have slight color difference due to light and display. Welcome to inquire, you can visit our company!
所有商品规格尺寸均可以根据客户需求定制，所以购买时需要提前联系预定；大件商品部分是没有库存的，如果需要，欢迎来电咨询或来厂洽谈！ Specification inventory: All product specifications and dimensions can be customized according to customer needs, so you need to contact the reservation in advance when purchasing; large parts are not in stock, if necessary, please call us for consultation or come to the factory to discuss!
Use of the oven:
1. The oven should be placed in a dry and horizontal place in the room to prevent vibration and corrosion;
2. Pay attention to the safe use of electricity, and install a sufficient capacity switch blade according to the power consumption of the oven. Use enough power wires and have a good ground wire;
3. The oven with an electric contact mercury thermometer type thermostat should connect the two wires of the electric contact thermometer to the two terminals on the top of the box. Another ordinary mercury thermometer is inserted into the exhaust valve. (The thermometer in the exhaust valve is used to calibrate the electrical contact mercury thermometer and the actual temperature in the observation box.) Open the hole of the exhaust valve. Adjust the mercury thermometer at the electrical contact to the required temperature and tighten the screws on the steel cap to achieve the purpose of constant temperature. But you must pay attention not to rotate the indicator iron outside the scale when adjusting;
4. When all preparations are ready, put the sample in the oven, and then connect and turn on the power. The red indicator light indicates that the box has been heated. When the temperature reaches the controlled temperature, the red light turns off and the green light turns on, and the constant temperature starts. In order to prevent temperature control failure, you must also take care;
5. When placing the samples, pay attention not to arrange them too densely. No test article should be placed on the heat sink to avoid affecting the upward flow of hot air. Baking of flammable, explosive, volatile and corrosive items is prohibited;
6. When you need to observe the samples in the working room, you can open the outer box door and observe through the glass door. However, it is better to open the door as few as possible, so as not to affect the constant temperature. Especially when working above 200 ℃, opening the door of the box may cause the glass door to suddenly cool and break;
7. In the drying oven with blower, the blower must be turned on during the process of heating and constant temperature, otherwise it will affect the uniformity of the temperature of the working room and damage the heating element. 8. After work is completed, the power should be cut off in time to ensure safety;
8. Keep the inside and outside of the oven clean;
9. During use, the temperature should not exceed the operating temperature of the oven;
10. In order to prevent burns, special tools should be used when taking and placing the test samples.